The sleeve is connecting
The sleeve is connecting, (an analog of a sleeve 55004) it is applied at measurement of pressure of environments with a temperature which is exceeding the limit of the operating range of temperature of a measuring device. For creation of normal temperature conditions connection of a membrane divider with the measuring device has to be carried out or through a connecting sleeve, or through the bringing tube which is established by the consumer between a point of pressure tap-off and a divider.
Pressure converters measuring pneumatic GSP always connect to a divider by means of a sleeve.
Shift when mounting a divider with a connecting sleeve on height is allowed, at the same time it is necessary to consider an error of the installation of a measuring device with an upper limit of measurement to 1 MPa determined by the hydraulic pressure of a column of dividing liquid in a connecting sleeve.
|Maximum pressure||60 MPas.|
|The standard sleeve connecting in the unrolled state has length||2,5 meters.|
|Material||stainless steel 12X18H10T.|
By the separate order the sleeve of any length, but no more than 5,0 meters is made.
- Connecting plug
- The connecting plug with an external thread of M20x1,5
- Connecting M20x1,5 union
- Capillary tube
- Nut of M20x1,5
For more information contact our sales department by phone. +7(495) 255-22-96, +7(495) 781-88-63.
We also recommend